We use Pressure controller Pressure detection is actually achieved through a measurement system. In order to measure accurately, we must do something right. Pressure controller For proper selection and calibration, attention should also be paid to the correct installation of the whole system. If only Pressure controller Its own display value does not fully represent the actual parameters of the measured medium.
However, the correct installation of the system includes the opening position of the intake port, the reasonable laying of the connecting pipe and the correct installation position of the instrument. Xiaobian introduces the correct installation method of the pressure outlet for the big * * * * * *, which is mainly divided into the following steps:
1, the location of the outlet should not be installed in the area where the pipe is bent, diverge and the flow of the beam forms vortex.
2, if there are prominent objects in the pipeline, the pressure outlets should be installed outside.
3, if it is installed in a wider container, the pressure outlet should be placed in a smooth and vortex free area.
4, if the pressure is adjusted near the valve, the pressure outlet is in front of it, then the distance between the pressure port and the valve should not be 2 times the diameter of the light rain. If the pressure port is in the rear, then the distance from the valve should not be less than 3 times the diameter of the pipe.
Changzhou Tianli Technical personnel are knowledgeable, welcome to discuss all kinds of technical problems, contact telephone number: 13656120670.